Fat Burning Frequency Therapy Machine
I. Treatment mechanism
HXY-AI model transcranial magnetic encephalopathy physiological therapeutic apparatus can be controlled by way of buttons. It is a comprehensive encephalopathy treatment device that stimulates limbs with electricity, based on brain physiology, biophysics, clinical encephalopathy therapeutics, electrical stimulation theory, and low frequency spectrum alteration treatment technique. There are various treatment frequencies and treatment strengths stored in the host, which can also be enlarged and upgraded. You can choose an appropriate frequency according to different diseases in clinical treatment. You can also choose various frequencies and output them alternatively in treatment. The host can display the information of 6 output roads in treatment simultaneously. Through treatment electrodes, the device can work on different points, sends out alternating current stimulation containing certain energy, which can directly penetrate the skull to reach the deeper tissue inside brain, acting on brain cells and central nervous system. As a result, the cells will get much more electricity, have a greater ability of carrying oxygen. Thus, the metabolic environment of brain cells will be improved, so will the activity of metabolizing enzymes, and there will be a higher possibility for harmed cells to be repaired. It will interfere and control the occurrence and transform of abnormal EEG and neuromagnetic, increasing brain electrical activity, promoting the recovery of brain functions. In this way, it will treat certain brain disease, and improve the sub-health condition of brain.
II. Product characteristics
This product is appropriate for all kinds of medical establishments. The host has 6 output roads, special fastigial nucleus stimulation electrode plates, and limbs electrical stimulation plates. It works on various nervous functional domains. The host is completely controlled by computer, easy to operate and easy to choose functions. The treatment electrode plates are convenient and comfort to wear. The lower treatment body is flexible for different treatment positions. This is nature therapy, safe and reliable, with no side effects.
III. Main technical parameters
1. Number of treatment channels
6 roads and every road can work independently.
2. Normal working conditions
a. Environment temperature: 5°C-40°C.
b. Atmospheric pressure: 70Kpa-406kpa.
c. Relative humidity: <=80%.
3. Fastigial nucleus electrical stimulation
a. Impulse brightness: 0.6ms±1ms.
b. Output wave types: 6 types.
Continuous wave (1kind) Dilatational wave (1 kind)
Light hammer wave (2 kinds) Massage wave (2 kinds)
c. Impulse range: 45V±10% (500 Ω load)
d. Period range: 0.01-1.99ms.
e. Frequency: 100Hz-0.5HZ.
f. Power source: 220V±22V, 50Hz±1Hz.
4. Electrical stimulation on limbs
Working frequency of the therapeutic apparatus is 2KHz-6KHz, and the allowable error is ±10%.
4.1 output current
4. 1. 1 the output current is continuously adjustable within 0~100mA.
4. 1. 2 the stability of output current is not larger than ±5%.
4. 2 on adjusting the frequency range
The modulation frequency range of the apparatus is not narrower than 0-150Hz.
4.3 diathermanous working frequency
The diathermanous working frequency is 2Hz, and the allowable error is ±10%.
4.4 surface temperature of the electrodes
The surface temperature of the electrodes is between 38°C and 50°C (below 23°C±2°C) in the process of diathermy.
4.5 Output wave form
Output wave form: Sine wave, square wave, triangle wave, exponential wave, peaked wave, sawtooth wave, continuous wave, dense wave, sparse wave.
IV. How to operate the machine
1. Insert one end of the special power line into the city power three holes specialized socket, and another end into the power socket on rear panel at the back of the host.
2. plug the pin that outputs electricity stimulation respectively into the output holes on the back board of fastigial nucleus, limb 1, and limb 2. Connect the electrode plates well.
3. Firstly, clean the treatment position with clean water or alcohol. Take one layer of non-woven fabrics and soak it (shake the droplets away). Next, stick the soaked fabrics on the treatment position and then stick the black side of the electrode plates directly on the non-woven fabrics. Fix it with strong bondage or keep it under control with sand bag. When you use it at home, you can also clean the treatment skin and the black side of the electrode plates with gauze having been immersed in hot water. Let the black side touch the skin directly.
Note: it is prohibited to pull out or insert the treatment cap or the electric stimulation pins when the therapeutic apparatus is powered on.
4. on how to operate the host
4.1 Fastigatum: press the mode button to choose from the 6 modes; every indicator lamp represents one mode. Press the period button to choose period, digitally displayed 0~99, adjustable. Press the start-key once, the machine starts working. The digital tube displays zero and a red dot beside it will shine. Now, press the increase key or decrease key to adjust the intensity within 0~99 and the period can also be adjusted at this time. The red dot will not shine when the digital tube displays period, but it will shine when displaying output intensity. Press the start/stop key again to stop output.
4. 2 Treatment of limbs
Press the intermediate frequency increase key or decrease key to choose prescriptions 1~35, adjustable. When you have chosen the right prescription, press the confirmation key, and the middle window displays "0", along with shining red dot. Now the treatment starts. The intensity can be adjusted through "" and "". The intensity shall be appropriate to the patient's condition (generally it shall be tolerable to him or her; otherwise it shall follow the doctor's advice).
4. 3 Diathermy function
When you need the diathermy function, you can press "+" or "-" at any time to adjust temperature. The 6 green lamp indicators at the front of the device represents 1~6 grades from left to right. The higher the grade, is, the higher the temperature.
End of treatment
The treatment program is designed to stop by itself and give a continuous warning sound at the end of treatment, but you do not have to operate it but just take down the electrode plates.
If you have to stop in the middle of treatment, just press the "cancel" key to stop treatment (including diathermy output). The digital tube displays the number of the prescription used just now. Diathermy indicator lamp goes out. You can take down the electrode plates or select another prescription to start treatment again.
Generally, it takes 20 min for every treatment, once or twice a day, one course two weeks. There shall be a time interval of 2~3 days between every two courses.
V. Matters need attention
1. Do not let the treatment table be too close to eyes, lest the electric stimulation hurts the eyes.
2. When receiving the treatment, you had better close your eyes, shut your mouth, do not watch TV or read newspapers so that the treatment effect will be better.
3. If such symptoms as headache, dizziness, fatigue, palpitations, shortness of breath, sweating occur in the process of treatment, it is because the output energy is overlarge. Just appropriately decrease the output energy and these symptoms will disappear. When necessary, the symptoms will disappear when you stop treatment temporarily.
4. You must take down the electrodes firstly at the end of treatment, and then shut the power source, for fear of electric shock.
5. When using the diathermy function, you must control the temperature, for fear of scald.
1. Patients of acute systemic and intracranial bleeding disorders;
2. Patients of intracranial infection, intracranial malignancy;
3. Patients installed with heart metal film and cardiac pacemakers.
VII. Repair and maintenance of common faults
1. The power source indicator does not light up
1. 1 the power source is not connected---check whether the power source is connected;
1. 2 the power insurance or the indicator goes wrong---contact the manufacturer for maintenance;
2. Treatment indicator lamp does not light up---treatment cap and host is not connected; check whether the output plug is connected firmly with the output hole.
3. Avoid no-load operation; avoid contacting corrosive chemicals.
Clean and disinfect the electrode plates.
After usage, clean and disinfect the electrode plates with alcoholic cotton ball.
Clean the dust; Keep the host and the table body clean;
Clean the surface of the device regularly with clean cotton cloth regularly (at least once every month).
VIII. Storage and transport
1. Refer to the regulations for transportation conditions and requirements;
2. Packaged products shall be stored in clean room with good ventilation, no corrosive gas, a relative humidity less than 93%, temperature -40ºC~+55ºC.
3. External dimensions of the host: 525×460×860.
IX. After sales service commitment
We provide one year warranty for the device after it has been sold, and we provide maintenance throughout its service life. Maintenance beyond the warranty period only charges costs of parts.
If you meet up with any problem when using this therapeutic apparatus or need to consult our company, please call the after sales service telephone number.
Registration and production address: Xihuan industrial park, Neihuang county, Anyang city, Henan province.
Tel.: 0372-7775555 refer to 1-12
Post code: 456300
|3||Electrode plates||8 pairs|
|4||Self-adhesive electrodes||4 pairs|
Fat Burning Frequency Therapy Machine