MCS-Argon-2000Y argon plasma coagulation,diathermy argon
MCS-Argon-2000Y surgical bipolar coagulator and cutter
LED High Frequency Argon Electrosurgical Unit
Technical Specifications of plasma coagulation:
Power: 220V±22V, 50Hz±1Hz (110V±11V, 60Hz)
Operating frequency: 512KHz
Power rating: 880VA±10%
Output Rating of diathermy argon:
a) Pure cut: 1W~300W( Load 800Ω)
b) Blend1: 1W~200W(Load 800Ω)
c) Blend2: 1W~150W(Load 800Ω)
d) Spray Coag: 1W~80W( Load 800Ω)
e) Forced Coag: 1W~120W( Load 800Ω)
Argon beam coagulator parameter
Flow rate Range 0.1~15.0L/min
Regulating Step 0.1L/min
Pressure of Nozzle 0-0.1MPa
Power Consumption <100VA
Argon Purity ≥99.99%
Pressure of Inputting 0.20-0.50MPa
AC Power AC90~250V, 50Hz
Configuration Sheet of diathermy argon :
Electrosurgical Pencil 5pcs
Electrosurgical Pad 10pcs
Electrosurgical Pad Cable 1pc
Bipolar Forceps and cable 1set
Argon Pencil 1pc
Laparoscopic Electrode 1pc
Main Parameters of argon plasma coagulation:
1. Humanistic board design, simple operation, direct-viewing demonstration.
2. Argon Plasma Coagulator and Electrosurgical Generator that unify the collection, has strong practicality and applicability.
3. REM system and Power Peak System that provides surgeons with improved performance at Lower power settings, minimizing the risk of tissue damage.
4. The newest AIC technology, double pattern argon current regulation, may respectively output 0.2-3.0L/min and 1-15L/min current capacity, the adjustment length of stride respectively is 0.2L/Min and 1L/Min
5. We use the SMART INTERRUPT safe system to real-time monitor nozzle pressure, which avoids nozzle jammed.
6. It initiates AUTO-ADJ. Adaptive pressure adjustment, allowing 0.20-0.50MPa broad argon indefinite pressure inputting, avoiding the limitation of the on-the-spot source of the gas.
7. It offers the effective and economical method for hemostasis and tissue deactivating in the opening digesting-scope operations.